VANTAKOOL Digital PH Meter, 0.01 PH High Accuracy
- Wide Use: Ideal for household or laboratory use; Suitable for testing pH balance of drinking water, pool, aquarium, RO system, Spa, or hydroponics; Appropriate for wide range of applications
- Fast And Accurate: Ensure your pool, general hydroponics ph or even drinking water has a perfect pH balance and safe for use with VANTAKOOL Tester precision pH accuracy. VANTAKOOL Tester has a sensitive electrode sensor to read accurate pH levels. The included detailed manual will show you how to get the best accurate and stable reading
- Auto Calibration: The VANTAKOOL PH meter has an auto temprature compensation feature that adjusts to water temperatures and water quality to give pH readings almost instantly. As well, you can calibrate the tester with one touch button using VANTAKOOL Buffer Powder.
- Lightweight And Easy To Use: Just remove the protective cap and immerse the PH meter electrode in the solution and press the button to be measured. Beats the heck out of trying to see the colors of a litmus strip.
Digital PH Meter Water Tester
This pH meter is affordable and useful for those who need a PH tester for household and laboratory use.
It’s not only ideal for household or laboratory use.
Also suitable for testing pH balance of drinking water, pool, aquarium, food Project, RO system, Spa, or hydroponics.
► Measuring range: 0.00-14.00 pH
► Resolution: 0.01 pH
► Accuracy: 0.01 pH
► Batteries: 2X 1.5V LR44 (included)
► Automatic temperature compensation: 0-50℃(32-122℉)
► Dimensions: 15×2.5×1.3cm/6×1.1×0.5 inch
► Net Weight: 52g/1.83 oz
▼ STEP 1: Turn open pH meter, Clean the electrode in distill water
▼ STEP 2: Immerse the electrode in pH 6.86 standard buffer solution
▼ STEP 3: Press the “CAL” button. Wait until the display digital is 6.86pH and steady
▼ STEP 4: Clean the electrode in distill water and dry it, immerse the electrode in Ph 4.00 standard buffer solution
▼ STEP 5: Press the “CAL” button. Wait until the display digital is 4.00pH. Then complete pH 4.00 calibration
▼ STEP 6: Clean electrode and dry it, put the meter into PH 9.18 standard buffer solution. Check if reading is correct between 9.17-9.19pH
► 1 x PH Meter
► 3 x Calibration packets
► 1 x Carrying case
► 1 x User’s manual